Bit 2 must be programmed high to use pin 3 as an input. The Action VIs – initiate or terminate test and measurement operations, such as arming the trigger system or generating a stimulus. Two types of measurement commands areavailable: The output mode switch is used to connect ordisconnect the the output capacitor located inside the unit. LEVelThis command selects the next point in the calibration sequence.
|Date Added:||13 October 2018|
|File Size:||18.38 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Agilent B Labview programming | Keysight Community
EITHertriggering occurs on either edge. The band limit above and below the trigger level is one half of thehysteresis value added to or subtracted from the trigger level. Once the measurement is complete, the Labgiew VI is called which performs an instrument error query and terminates the software connection to the instrument.
Pin 3 is controlled by bit 2, and can be programmed to serve either as an input or anoutput. C – Error MessagesTable C The second parameter is the password.
Themeasurement trigger system is described under “Triggering Measurements”. This increases the low current measurement sensitivity forgreater accuracy and resolution.
It is also recommended that agipent set Disable Auto Serial Polling. Nonvolatile memory has a finite maximum number of write cycles. Type to filter by text Filter by tag Sort Sort by date created: The Utility VIs might also include other custom instrument driver VIs that perform operations such as calibration or storage and recall of setups. The Action VIs – initiate or terminate test and measurement operations, such as arming the trigger system or generating a stimulus.
The instrument first measures theminimum and maximum data points of the pulse waveform. Measure commands measure the output voltage or current.
How to Use an Instrument Driver in LabVIEW Tutorial – National Instruments
Themeasurement capabilities of the Agilent A 66632b Agilent A models are particulary useful forloads that draw current in pulses.
CError MessagesError Number ListThis appendix gives the error numbers and descriptions that are returned by the dc source. Show 0 Likes 0. This information is saved in non-volatilememory.
Get a feed of this content Use this view in a tile. This protection feature is implemented with the VOLTage: Display commands control the front panel display of the. You cannot measure output voltage and current simultaneously.
How to Use an Instrument Driver in LabVIEW Tutorial
I only see two modes Constant Current mode and Constant voltage mode. This tool searches IDNet to find the specified instrument driver.
Defines the technical details of the GPIB interface. Programming the DC Source – 3Trigger occurs on rising edgeTrigger occurs on falling edgewhen signal crosses positivewhen signal crosses negativehysteresis band limithysteresis band limitTRIG: As shown in figureyou can move the block of data being read into the acquisitionbuffer with reference to the acquisition trigger.
POSitivetriggering occurs on the rising edge. The followingfunctions can be selected: The returned value is 15, if the switch is setto Normal and 60, if the switch is set to Fast.
They are defined by the IEEE Pin 4 isthe digital ground. I am using a NA Power Supply. Refer to these for all non-SCPI commands for example: