Agilent HP A 6. HP A current source for resistance function test. This application note discusses the compatibility and differences between the Keysight A and A Digital Multimeters. Overall signal path is rather simple to read on board, even without reference to service manual or schematics. Declaration of Conformity Search Keysight Regulatory database for the most recent Declaration of Conformity statement for your product. Few photos of calibration steps for better understanding.
|Date Added:||5 October 2008|
|File Size:||32.65 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
DMM is very dirty outside, and definitely need good deal of scrubbing with alcohol and cleaning. After calibration, values are compared to Keithley meter to ensure they agree on readings. HP A current source for resistance function test. Hp Agilent Keysight A.
Agilent A: Multimeters | eBay
When you use a digital multimeter to measure AC voltages, it is important to know how your DMM makes the 344401a and understand your instrument’s limitations.
All buttons, knobs, and switches are good and intact. This application note is the second in a series of three, that will help you eliminate potential measurement errors and achieve the greatest accuracy with a DMM.
Not this time though, no problems detected with a switch, probed input 1. Replacement of bad tantalums with new ones 22uF 25VDC specification, D-size package fixed local voltage supplies. Showing the additional capabilities of the A DMM.
Front panel board REV. The serial is SG Top Dog Test is corporate friendly we understand the traditional acquisition and capital approval process.
Providing the finest test equipment solutions since 1992
If you like the article or have any questions, jump in comments! We tested this item and it appears to be functioning properly, though we did not run it through a Layout is tidy and nicely done for this instrument, thumbs up to HP engineers.
LM is socketed, just like more recent 34401aa instruments. See the new graphical user interface, display and measurement capabilities.
| HP A repair and calibration worklog
Data Logging and Digitizing Using a Digital Multimeter This measurement tip describes the use of a digital multimeter for data logging and digitizing. Calibration was referenced to measurement results of my Keithley which was calibrated in February by Tektronix, and is most accurate meter I currently have up 34401aa date Few photos of calibration steps for better understanding.
Up for sale is a used, complete, sold as pictured, no errors, passed self test but will need a new display as there are extra lines and or digits in the display and unit will be sold in working order HP A local 15V zener regulator. I had some left from other repairs, so you can see already new caps on photo above.
This measurement brief describes how to use a DMM to make multisignal captures and analysis without programming. See search results instead: This part is used and was removed in working condition. Agilent A Digital Multimeter, 6.
HP A front panel removed, front and rear. This application note discusses the compatibility and differences between the Keysight A and A Digital Multimeters.
34401A Digital Multimeter, 6½ Digit
Redistribution and use of this article or any images or files referenced in it, in source and binary forms, with or without modification, are permitted provided that the following conditions are met:. And yes, some were dead short, and some were missing, likely due previous owner repair attempts.
Some of hobbyists were even upgrading reference to LTZ circuit for better A stability. This measurement tip describes the use of a digital multimeter for data logging and digitizing. First step — zero calibration with 4W short. Final test with 10VDC: